For Blood Purifier, there are two “ideal” places to place the TENS pads. One is on your arm: one pad on your wrist, and the other further up your arm. In this way, your blood is exposed to the healing … Continued
MicroGen is a current output generator, so power levels vary greatly according to the load. All programs on high power can theoretically deliver 0.084 watts of continuous power. Low power delivers about 0.0075 watts into a 60k load.
You can choose multiple programs which can up to 1000 frequencies.
Yes, it is normal. When the power level flashes, it means the power level. When it back to be stable, it means the electric quantity
Octave harmonics are used. Lower frequencies use higher harmonics.
You can check the below link for details: http://www.spooky2tools.com/terrain-protocol-programs-for-heawea-microgen/?fbclid=IwAR2WMEUPd7KAIKq5i9dAv11bTtccyx4dlafqrkC8RKXwjGjeabfacUxQpk0
First of all, HeaWea MicroGen cannot do any biofeedback. But you can use Spooky2 products to perform biofeedback scanning, and input the results into HeaWea MicroGen software, create a program of your own, and then run it. You can wear … Continued
For Blood Purifier, the suggested power level is 3. For Zapper, the suggested power level is 1. For all other modes, the optimum power level should be set low, and gradually increased if there are no side effects.
You can try these steps: 1. Make sure your High power cable is fully plugged in. 2. Download the latest version of HeaWea software from www.heawea.com 3. Upgrade your firmware to be the latest one. If you do not know … Continued
You can always use the Low power cable with a High Power program, you will just get low power. If you use the High Power cable with a Low Power program, you may get uncomfortable or even painful shocks if … Continued